|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Packaging Details:||Paper box|
|Delivery Time:||7 work days|
|Supply Ability:||30 per month|
|Measuring range:||0-10mm, 5-15mm, 10-20mm, 20-30mm, 30-40mm||Resolution:||0.01mm|
|Accuracy:||0.03mm||Max. moving speed:||1.5m/s|
Digital Wedge Feeler Optical Linear Scale Electronic Display Feeler Gauge
Used for the static or dynamic gap measuring on the turbine or hydraulic turbine.
Changing battery : open the cover and change the battery, close the cover and the battery will be locked.
ZERO, reading set 0, in any position users can set as 0.
ON/OFF: power on / off
Before usage, loose the screw, move the scale and use clean rag to clean the scale sides. Move the reader head to the right end, makes the auxiliary scale combining with the wedge, set the reading 0.
We offer 5 models of the Wedge Feelers
UNIMETRO UXY-1 0-10mm
UNIMETRO UXY-2 5-15mm
UNIMETRO UXY-3 10-20mm
UNIMETRO UXY-4 20-30mm
UNIMETRO UXY-5 30-40mm
Users are able to select the model based on requirements.
Measuring: during the measuring process, the users need to insert the Wedge Feeler into the wedge groove, the bottom side of the auxiliary scale should be full contact to the measuring surface,and vertical to the bottom.
Direct measuring: model 1 wedge feeler (0-10mm) is able to achieve direct measuring, the LCD will display the measuring data from the wedge groove.
Adding measuring: model 2, model 3, model 4, model 5, users should align the auxiliary scale with the wedge block end side and set the position 0, than do the measuring.
The display value from the LCD + the reading value from the wedge (model 2 + 5mm, model 3 + 10mm, model 4 + 20mm, model 5 + 30mm) = the real measuring value
For example: the display value is 7.5mm, the wedge is marking +5mm (model 2), the real measuring value = 7.5 + 5 = 12.5mm
|Measuring range||0-10mm, 5-15mm, 10-20mm, 20-30mm, 30-40mm|
|Max. moving speed||1.5m/s|
|Customized is available, the delivery should be 15 work days|
1. Do not expose the products in the dusty or wet environment, the humidity should be lower than 80%, prevent the UV or high temperature radiation.
2. The moving scale block should be balance, prevent dropping or crashing.
3. Do not close to object with magnetic.
4. Use the gasoline to clear the dirt, use the clock lubricant for the sliding surface and the wedge, do not use alcohol or acetone. The electronics and the scale body prevent any liquid.
Normal error solution
|LCD does not display or the character flashing||Out of battery or the battery is loosen||Change or adjust the battery|
|The display runs chaos||The scale is moisture||Clean the scale with clean rag|